|Place of Origin:||Shanghai, China|
|Certification:||CE, ISO, SGS, FDA|
|Model Number:||500 LED|
|Minimum Order Quantity:||1 set|
|Packaging Details:||In two wooden case|
|Delivery Time:||5-7 days after the receipt of payment|
|Supply Ability:||500 sets/month|
|Illuminance Depth:||1300mm||Size Of Light Field:||160~280mm|
|Power Consumption:||60W||Total Bulb Quantity:||72pcs|
|Lowest Installation Height:||2900mm||Warranty:||2 Years|
60W adjustable illuminance plastic surgery used shadowless LED surgical light
1.LED series shadowless Operating lamp can widely be used in diversified operation occasion to meet the demand of the illumination, and it is the ideal illumination instrument of the modern operating theater.
2.The Whole reflection optical system is made by CAD/CAM technology, its illumination depth is up to 1300mm ;by using color temperature compensation technology, the color reversion is uprised, and it is more suitable for the illumination of the complex surgery such as encephalon surgery and thorax surgery.
3.The panel is made of German PC (anti-bullet plastic) by injection moulding, makes the light softer and more reliable,Equipped with imported balance arm suspension system, which is light and handy, six group joints move together and moves smoothly, positions stablely. 360 universal design can satisfy various needed height and angle in the operation.
4.Almost unlimited LED life expectancy delivers tremendous product reliability and investment protection.
How about the detailed specification?
|Color reduction index||96Ra|
|Size of light field||160~280mm|
|Lamp head diameter||500mm|
|Rated power of bulb||3.2V/1W|
|Service life of illuminant||50,000h|
|Power supply voltage||AC110-240v,50/60Hz|
|Total power consumption||60W|
|Total LED bulb quantity||72pcs|
|Lowest height of installation||2900mm|
1. Troubleshooting conducted by the user
|Power light on the Control panel is off||Power off||Switch on the power source on the wall surface|
|Wire on the Power panel is disconnected||Reconnect the wire on the power panel|
|Power light on the Control panel is off||Fuse is burnt out||Replace the fuse|
|Illuminant of operation shadow less lamp blinks||Power voltage over the network is too low||Regulated power supply shall be installed|
|Bulb indicator light on the Control panel blinks||The bulb is burn out||Replace the bulb|
|Over high or small equilibrant||Elasticity of the balancer is too high or too small||Adjust the elasticity of the balancer|
|Spinning body is unable to be positioned||Bed of the Spinning body is imbalanced||Adjust the bed nut|
|Damper is loosened||screw the damper tightly|
|Lamp body is unable to be positioned||Screw inside the rubber band of the lamp body is loosened||Whirl the screw tightly|
Contact Person: admin