|Place of Origin:||Shanghai,China|
|Minimum Order Quantity:||1 set|
|Packaging Details:||In three wooden cases|
|Delivery Time:||5~7 days after the receipt of payment|
|Payment Terms:||L/C, T/T, D/A, D/P, Western Union, MoneyGram|
|Supply Ability:||300 sets/month|
|CRI:||94Ra||Size Of Light Field:||160~280mm|
|Lamp Head Diameter:||600mm||Total Power Consumption:||70W|
single halogen bulb 60cm diameter over 120000lux illuminance halogen shadowless operating lamp
1. Position adjustment of lamp cap
As installed, power on the lamp cap, so that it will be lit up with time-delay. Pull the handle to make the lamp cap move from upside to downside, and vice versa as well as conduct the axial rotation. Push the sterilized handle, so the lamp cap could be inverted backwards & forwards (range is up to 200°) and the right & left (range is up to 200°). Finally, the lamp cap is adjusted as you desired.
2. Rotate the sterilized handle, so as to adjust the light spot as required.
3. Switch & intensity control
Lamp cap switch is controlled by the switching push button. As switched on, the lamp cap would be lit up with time-delay. As required by the operation, the intensity of light can be adjusted via luminance enhancement button or luminance weakening button. Here, the intensity of light will be displayed on the brightness record.
Push down the power switch button, the two-color power light turns into green, brightness indicator light will be displayed on the last brightness record left by the operation shadow less lamp. Also, the light spot brightness thereof is adjusted to the related level. The control system is provided with the soft startups program, which can be used for the protection and extension of the bulb’s life. Therefore, as starting up, it is normal when the intensity of light spot is changed from darkness to brightness with a couple of seconds.
Push the light changes button to adjust the intensity of the light spot, so as to comply with the doctor from the point of vision.
|Color reduction index||94Ra|
|Size of light field||160~280mm|
|Lamp head diameter||600mm|
|Rated power of bulb||24V/150W|
|Service life of illuminant||1,000h|
|Power supply voltage||AC110-240v,50/60Hz|
|Brightness adjustment||Automatic 8-stage continuous light adjustment|
|Total power consumption||70W|
|Total LED bulb quantity||1pcs|
|Lowest installation height||2900mm|
How about the package?
|Packing size||Single head:91*92*37cm
|Weight||Single head:GW:19KG NW:12KG
The enterprise product registration standard should be applied.
Fully apply the national standard, GB9706.1 Medical Electrical Equipment, Part I-General Requirements for Safety (equivalent to IEC 601-1, Medical Electrical Equipment, Part I-General Requirements for Safety)
Our product and the related instructions, advertising articles and samples were audited and approved by the authorities, and then be protected by the law concerned.
Transportation and Storage
This apparatus shall be transported and stored in the following environments:
1) Ambient temperature: -40°C~+55°C
2) Relative Humidity: £93%
3) Atmospheric Pressure: 500hPa~1060hPa
Contact Person: admin