|Place of Origin:||Shanghai,China|
|Minimum Order Quantity:||1 set|
|Delivery Time:||5~7 workdays after payment|
|Payment Terms:||L/C, T/T, Western Union|
|Supply Ability:||500 sets/month|
|Main Unit:||1 PCS||Double Foot Switch:||1 PCS|
|ESU Pencil:||1 PCS||Electrode (Disposable):||1 PCS|
|Bipolar Forceps:||1 PCS||Bipolar Forceps Cable:||1 PCS|
|Electrode Cable:||1 PCS||Power Cable:||1 PCS|
|Fuse:||2 PCS||Warranty:||1 Year|
Portable Surgical Diathermy Electric Knife Surgery/Electro Surgical Unit for ENDO surgery
Medical High Frequency Electrosurgical Uunit with isolated output design, it provides proper power outputfor cutting and coagulating tissue in both monopolar and bipolar s urgeries. It can meet the needs of many surgical operations. Adopting singlechip microcomputer control technology and modular design, it features simple design, convenient maintenance, powerful functions, stable performance and easy operation, and is equipped with versatile jacks for
1. Ambient temp. range: 5ºC～40 ºC
2. Relative Humidity Range: ≤80
3. Atmospheric Pressure Range: 86.0～106.0kpa
4. Power: 220V±22V, 50Hz±1
5. Operating Freq: 330～460kHz;
6. Rated Power: 1100VA±10%
Power Consumption: ≤1100VA (Cut function 400W)
Transportation and Storage
Ambient Temp Range
~40 ºC ～ 55 ºC
500hpa ～ 1060hpa
Nine output mode
a) Cut: 400W
b) Blend 1: 250W
c) Blend 2: 200W
d) Blend 3: 150W
e) Electro Coag 1: 150W
f) Electro Coag 2: 120W
g) Electro Coag 3: 100W
h) Bipolar Coag 1: 70W
i) Bipolar Coag 2: 70W
After well connect the cables for Electrosurgical unit & Endoscope;
Well install power cable, Electrode cable, Electrode.
Turn on electrosurgical unit, once it has data indication in display window, press “Endoscope” button, then the machine will enter into self adjustment status.
It will adjust to the high voltage cut technology, spray coagulation technology which suitable for operation under endoscope, which can be use for vaporization working, gastroscope cut, gastroscope coagulation, hysteroscope use etc.
matching the electrode contact inspection system, which real-time test and evaluate the effective contact area between electrode and derma, if the contact area was found low to a danger level, system will automatic cutout and gives alarm indication.
Independent output interface, ensured that only the opened interface can gives output, other closed interface has no output, which is more safe.